Yes. With those settings, you'll get the average sound level over 20 seconds in each whole or 1/3 octave band, plus an overall (full bandwidth) average level. You can see the overall level by tapping the graph to turn on the cursor.
This section allows you to view all posts made by this member. Note that you can only see posts made in areas you currently have access to.
The new SoundMeter apps also support the Dayton UMM-6 USB measurement microphone. When the UMM-6 is detected, the user will be prompted to enter its serial number, which is typically printed on a label on the UMM-6. If a valid serial number is entered, SoundMeter will automatically download a calibration file from Dayton’s website and set the microphone sensitivity based on the information in the file.
Note: The Dayton UMM-6 must be connected to your iOS device via Apple’s USB camera adapter (either for Lightning or 30-pin dock connector, depending on your device). Sometimes the UMM-6 shows up in iOS as USB PnP Sound Device. If that happens, unplugging the UMM-6 and plugging it back in will generally solve the problem (it may take 2 or 3 tries). When properly connected, “UMM-6″ will appear as the input device name in SoundMeter’s toolbar (at the top of the screen).